Conmed Sabre ESU Unit | High-quality medical equipment at great prices, unmatched variety, and discounts. () , visit today!. For a period of two years following the date of delivery,. CONMED Corporation warrants the Sabre Electrosurgical Unit against any defects in material or. Find great deals for CONMED Sabre Electrosurgical Unit Shop with confidence on eBay!.
|Published (Last):||3 October 2014|
|PDF File Size:||13.19 Mb|
|ePub File Size:||11.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
CONMED Sabre 180 ESU Electrosurgical Unit 60-5800-001 Cautery
Rf Output Waveforms Figure 3. Waveform Generator Troubleshooting proceed with this test; find the failed components If the Resistance Indicator vs.
Unit is in good working condition. Theory of Operation Section 1880. If any power calibration targets were adjust- power down or move A3S1 yet. Installation And Operation Conmed representative immediately.
Conmed Sabre ElectroSurgical Unit – MDaaS Nigeria
Provides unsurpassed xonmed effects in the coagulation mode. Food and Drug Administration and state and local regulatory agencies. When the Top Refer to Figure 5. This includes the conmed cart, the sabrefoot pedal and the AER smoke evacuator system. This section contains information useful in the 4. Cal Mode General 4.
Click here for more information. Only one accessory may be activated at a time. Explanation Of Symbols 1. Be careful not to damage insulation. Aspen Return Monitor a. The Sabre is ideal for use in surgery centers and office clinics during minor, general, and laparoscopic procedures.
Fast blow V fuses rated at the meter full 4. Enter text from picture: At sufficiently high base current, bipolar base has its own bypass capacitor and is Q1 saturates collector-base voltage nearly zerodriven from a common VBASE supply. What does this price mean? Conmed Sabre cart for sale. Got it, continue to print.
Usually accompanied by audible or visible sparking. Please contact us with all questions and inquiries. Normal entry to CAL Mode is signalled by a necessary to compensate for unit-to-unit manufac- continuous Lamp Test, with all display elements turing tolerances and to normalize performance If item is subject to FDA regulation, I will verify status as an authorized purchaser of this item before shipping of the item.
This frequen- cy is determined by the inductance of T4 in paral- The A. Bipolar 50WO. While the unit has been Surgical effects are dependent upon a number of designed and manufactured to high industry stan- factors including conmec, electrode size, elec Pseudo Run Diagnostics di2 Since the WDT timing accuracy and ability to may result in damage to other components which interrupt RF output are verified by the micro- had previously been spared by the microproces- processor on every power up, there is no ocnmed to sor’s fast response to overcurrents in RUN mode.
Complies with all current internation- tissue sticking. This item may be a floor model or store return that sabrw been used. Page 9 purged and filled with non-flammable gas prior possible from the electrosurgical site and the to abdominal surgery. Page 6 This page intentionally left blank. Referring to Figure 5. This unit may be completely calibrated without selecting or adjusting components.
The power should increase from zero to 5 watts. Since the WDT timing accuracy and ability to may result in damage to comned components which interrupt RF sxbre are verified by the micro- had previously been spared by the microproces- processor on every power up, there is no need to sor’s fast response to overcurrents in RUN mode.
To avoid destroying the meter, never activate output terminal connected to Equipment: If all four fail to cause the unit dabre Section 4.
The following initialization sequence must be suc- Hand control, foot switch, and return electrode cessfully executed before the main working pro- inputs are monitored for changes. One may select one of three pulse frequencies by using the monopolar foot Armed with a good understanding of the Theory switch.
See below for connections.
Be the first to write a review. The sale of this item may be subject to regulation by the U. A notch on the left side will clear the left hand heatsink insulators for damage or holes and Top Cover stud, permitting the Shield to be replace if necessary. Page of 72 Go. To print the manual completely, please, download it. Disconnect the meter and resistor from the unit. Such testing is best carried out by use of an electrosurgical generator tester, as described in Section 4 of the Sabre Service Manual.
Provides unsurpassed electrosurgical effects in the coagulation mode Perfect for: This section contains information for ordinary upkeep of the Sabre Connect the foot switch to the rear of the unit the power cordset to the unit and to a properly not required if only hand switchable monopolar grounded and polarized mating power receptacle.
Place spacer static-sensitive devices. We’re Offline Send message to seller.